|Place of Origin:||Zhejiang , China (Mainland)||نام تجاری:||YOMIN|
|Voltage:||110V , 220V , 380V , 415V||Model Numbe:||DT - 2F96|
|Accuracy Class:||1.5||Operating Temperature:||- 25℃ ~ + 40℃|
|Measuring Range:||45Hz-65Hz||Dimensions:||96 * 96 , 72 * 72|
96x96 45-65Hz Analogue Panel Meters / Dual Frequency Meter
Panel meter is one kind of electric measure indicators with the direct effecting of imitative display. It is suitable for AC and DC output and input distribution circuit, power plant console cabinet, power operating board and different apparatus. It is used to measure DC current, voltage and AC current, voltage, frequency and phase power.
Frequency:45-55HZ 45-65HZ 55-65HZ 47-53HZ 57-63HZ
Frequency meter is the instrument which measures the frequency. Frequency meter shows the number of the cycles in 1 second. Frequency meters are connected parallel to circuit which frequency will be measured like voltmeters.
|Model||Frame size||Measuring Range||Class Index||Case Qty.|
Note: Please mention the voltage when ordering, our standard voltage are 110V,220V,380V,415V, special voltage upon request.
Scale Plate :
The normal scale plate is made of alloy-aluminium material white printed with black scales,Values, units, technique symbols, etc.
Scale plate with special requirements are available upon request.
Such as color, scale with lighting functions and PVC noctilucence are available upon request.
Black base and frame , white scale plate and meter body.
Normaly, the pointer is made of alloy-aluminium, it is black, and red, white are available Upon request.
All meters are subject to anti-static treatment to minimize the static-effect to the accuracy of the meters.
Operating position :
Generally, meter is installed in vertical position. Other installation position can be customized (please specify the desired angle).The accuracy class is just valid for the Corresponding installation mode.
Insulation Test :
The test Voltage for BE & BP series is 2KV RMS for 1 minute with the frequency 45~65Hz.
Safety Precautions :
Meters with damaged bezels or glass faceplates must be disconnected from the mains. Scale plate, front frames and windows may only be replaced under voltage-free conditions.
The meter should be kept clean, if the meter is used in a manner not specified by the manufacturer, the protective function may be impaired.
تماس با شخص: paola